As it states in the KB its a good idea changing the default timeout period to more than ms. Tektronix tkafg3k Signal Generator. Much easier to add to existing code. You can send up to 10 dealer emails at a time, and up to 30 per day. Have you run the functions by themselves?
|Date Added:||8 November 2013|
|File Size:||5.12 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Program will run better and you should be able lbview avoid glitches. Dual channel models can generate two identical or completely different signals. If you have not downloaded NI Package Manager, download it now.
It’s got two channels. I have a few more questions Shortcut keys provide direct access to frequently used functions and parameters. A large screen shows all relevant waveform parameters and graphical afg3021bb at a single glance.
It looks like everytime you change a setting on the AFG you open and close your communications. Based on your description of overall system, I think that you’re on the right track.
I’m testing the AFG alone to start. Share or Embed Document. I am currently considering purchasing the AFGB. The Dealer Directory is intended for visitors to make specific business inquiries and ,abview not be used to send bulk emails.
Afgb labview download
Would you like to receive announcements about auctions and special sales? This reduces the time needed for learning and re-learning how to use the instrument. At some point, you should address the errors that you get. Dual channel models can generate two identical or completely different signals. Notice that this is a different utility than VI Package Manager.
Overview of Instrument Drivers An instrument driver is a set of software routines that control a. FSK labview datasheet, cross reference, circuit and application notes in pdf format.
This gives full confidence in the signal settings and lets you focus on the task at hand.
Tektronix AFGB Used and New Function Generators –
Message 3 of Sorry, I haven’t made it clear but I’m communicating with the instrument via usb. I have cheated my way around this problem for now – see attachment – but I would still appreciate some help. Familiarize yourself with the printed operation manuals for the TDSB oscilloscope. I’ve been modifying some of the examples provided and all seems to be well. I have attached 4 zip file in LabView version 8. That’s the way the driver was intended to be used.
Version download access trial free Familiarize yourself with the printed operation manuals for the TDSB oscilloscope. As far as I can tell it is doing what I want it to do. I tried to reproduce the issue with AFG with firmware 3.
The problem Albview have occurs when I do not want the instrument to Reset on Initialize. Message 1 of Any further help much appreciated, as this is taking far too long and it should be easy! You could very well be causing instrument errors.